- Install the Pickering driver software
- Install the hardware
Disk-based installInsert the DVD, it should automatically display the following page in your browser. If this does not happen, start a browser and open the file 'index.html' on the DVD.
Next click on the Drivers heading to arrive at this page:
Then select PCI and PXI Switch Cards (System 40, 45 and 50) to arrive at the driver selection diagram:
Click on either the Direct IO driver or the VISA driver. Currently, both drivers are contained in the same installation file and selection if driver is made during the following installation process.
NOTE: If the card is to be used in a National Instruments software environment then in the vast majority of cases the VISA driver should be selected during the installation process.
Click on pilpxi/pipx40 to start the installation of the driver.
Go to: http://www.pickeringtest.info/downloads/drivers/PXI_Drivers/
Select Pipx40vpp-31.zip to download.
Extract the file to any suitable location on your PC and then execute the file setup.exe from that folder.
During the installation process, you will be offered the choice of a VISA or a Non-VISA driver. If you wish to use the VISA driver you will need to install National Instruments VISA before installing the Pickering driver.
Turn off the power to your PCI or PXI hardware system and install the Pickering card.
Re-start the power. In the case of a PXI system, the PXI rack should be turned on before the computer system to ensure that the PCI bus and cards are powered before the PC BIOS starts.
During the initial power-up sequence, Windows should detect the presence of the PCI/PXI card and install the software drivers. If this fails to happen, check your procedure very carefully; make sure the drivers were installed and the correct power-up sequence was used.
Run the test panel provided in the installation folder.
This panel requires the LabView 8.5 run-time engine to be present. If it is not already installed, a copy of the engine can be obtained from your National Instruments software source, or a copy of the engine is available on the Pickering distribution disk.
D:\PI_DVD\DVD_Contents\Run Time Engines\LabVIEW\LVruntime8.5
The Test Panel will list all cards found in the PCI/PXI system.
Click on an individual card to open a panel allowing individual relays to be operated.